Incinerator should be smokeless, odourless combustion and it should be made by high-quality cast, insulation, and steel plate as well as minimum generation of dust. Moreover the incinerator should be corrosion resistant.
Type of incinerator : Pyrolitic – hot medical waste disposing machine
Operation Condition: 8-16 hr/ day
Control: Built-in data recording
Operating temperature: Incinerator
/Primary Combustion Chamber
Primary Chamber: 900 – 1200 0C
Type: continuous loading, Top Loading
Capacity/Burn rate per hour: 250 -300 kg/hr
External- 3 layers
Internal lining: a fire proof material of pre-fired refractory bricks with Aluminium lining, resistant to corrosive waste or gas and to thermal shock
Temperature: 1200- 1300 oC
Residence time of gases : >2 seconds
Ash Residue: <5% of original waste size
Ash Handling System: Both Automatic and manual removal of Ash. Must ensure removal/treatment of hazardous remnants of ash
Flue gas treatment system : Capable of treating the flow of flue gas as the incinerator is operating at its maximum capacity
Auxiliary device: Water level gauge, pressure sensor, PH sensor..etc
Auxiliary device: Fuel cut-off device
Waste feeding mechanism: Automatic pneumatic/hydraulic waste loading system or conveyor belt , capacity 650-800 L at a time
Type: Vertical type
Material: Fireproof cast, stainless steel
ASH -Max <5% of original waste size
Reduction of Pollutant gas SO2, HCL, HF and line particulate
Test report for emission testing provided?
Heat exchange mode: Automatic
All standard accessories for incinerator, including but not limited to loading system, heat exchangers, pollution control system, ash removal system, including ladder and oil tanker (2500litre capacity).
The incinerator is capable to operate at the altitude of 2400mt above sea level. (according to the site conditions)
220 Vac single phase or 380 Vac three phase 4 wire system 50HZ
Installation Testing and Commissioning : to be conducted by certified or qualified personnel.
Supplier shall provide the following documentation
User (Operating) manual in English.
Service (Technical / Maintenance) manual in English.
Certificate of calibration and inspection from factory.
Fast moving spare parts: Supplier is able to provide fast moving spare parts with quantities as described in the price schedule or their equivalent.
Training: Supplier is able to provide training on operation, management and maintenance of incinerators.
Comprehensive warranty for minimum 2 year.
Maintenance Service during Warranty Period
During warranty period supplier must ensure, corrective/breakdown maintenance whenever required.
Supplier has a local agent or branch office in Ethiopia (please indicate the name and contact details of the agent)
Supplier provides labor, lifting equipment and tools needed for the installation, testing and commissioning of the incinerator. The supplier will be responsible for the safety of its staff, workers and laborers during the installation, testing & commisssioning
Able to provides all necessary information that would be used as an input for preparation of floor (platform) and room for the incinerator to be supplied such as the following;
Lay outs (drawings) and pictures of the incinerators to be installed
Length, width and height of incinerator,
Area (length x width) for the floor (platform) and Length, width and height for roofing, for each type of incinerators to be installed
Needs during the transport, installation, assembly, commissioning and operation of the equipment in terms of access points, available space to operate (inside and outside of the room), ways to get to the site where the equipment will be placed or any others. ;
Technical Specifications (fuel consumption rate, weight, anchoring system, support points, etc.);
Chimney specifications to be used for the roof design ( size, support needs, insulation);
Considerations related to operation and Maintenance, for e.g., minimum available area to perform routine maintenance tasks and replacement of key parts that need to be changed more frequently;
Any consideration/need of the incinerator while being operated that might impact the design of the room/shelter where it will be placed.
Other important information to be considered that can influence or impact the designing and building processes of the infrastructure for the incinerators, for example area dimension for other accessary parts like fuel tanker storage, if applicable, etc….
Delivery time (DAP, Addis Ababa) in weeks
Name of manufacturer including website of the manufacturer &/ supplier & country of origin
Thermocouples (2 pcs /each thermocouple assembled on the unit )
Thermocouples with pocket (1 pcs/each burner )
Nozzles (2 pcs/burner)
Nozzle line (1 pc/burner)
Glass fibre (2 roll/size)
Ignition electrode ( 2 pcs/burner)
High tension cable ( 2 pcs/burner)
Flame photometer (1 pcs/burner)
Burner primary chamber (1 set)
Burner secondary chamber (1set)
Control panel parts (1 set)
Fuse (1 pk from each rating )
Solenoid valve (1 pcs/assembled on the unit)
Burner oil filter (10 pcs/burner)
Fuel pump (1 pcs/burner)
Shaft seal kit (1 set /shaft)
Seal plate gasket ( 1 set / plate )
Mechanical seal (1 set)
Flue gas restricting ring bricks (1 set)
Refractory cement (1 pk )
Shaft bearing (1 set)
V belts (1 set)